|The device is not working when it is turned on, the alarm condition (red light) is activated and the power indicator (green light) is not illuminated.||1.The power cord plug is not properly inserted into the electrical outlet.|
2.No power from the electrical outlet.
3.Internal part failure.
|1.Make sure the device is properly plugged into the electrical outlet.|
2.Check your household outlet fuse or circuit breaker.
3.Contact Air Products.
|Lower oxygen (yellow light) activated.||1.The air flow to the device is impeded or blocked.|
2.The flowmeter knob is completely closed.
3.The oxygen tubing is kinked and blocking the delivery of oxygen.
|1.Check the condition of the air inlet filter and clean or replace as necessary. Remove any items that appear to be blocking the air flow into the device.|
2.Turn the flowmeter counterclockwise to center ball on prescribed flow rate.
3.Check to see that the tubing is not kinked or blocked. Replace if necessary.
|Alarm condition (red light) activated, when the device is on with the power indicator (green light) on.||Internal system failure.||Contact your Air Products technician immediately.|
|Limited oxygen flow to the user.||1.Faulty oxygen tubing, cannula or face mask.|
2.Faulty humidifier bottle.
|1.Inspect and replace the items if necessary.|
2.Inspect and replace the item if necessary.
|Alarm is weak or not sounding when the power cord is unplugged and power switch is on.||The 9V battery is weak.||Contact your Air Products technician.|
Don’t Be A Stranger…
Just write down some details and our customer success heroes will get back to you in a jiffy!