How long is the service life of an oxygen-producing molecular sieve? How often does the molecular sieve of the oxygen generator be replaced? How to extend its service life?
The molecular sieve oxygen generator uses air as raw material, the air is separated after multi-stage purification, and the oxygen produced is filtered and disinfected before being fed into the central oxygen supply pipe network, in addition, the biggest consumption of molecular sieve oxygen generator comes from the machine itself and the electric power. It can be switched on at any time without transportation. Replacement time of molecular sieve of an oxygen generator and how to prolong the life of medical molecular sieve oxygen generator.
Why do most people ask about the lifetime of molecular sieve when buying a molecular sieve oxygen generator? This is because the molecular sieve and compressor are the two most important components of the molecular sieve oxygen generator. The role of molecular sieve is to separate oxygen and nitrogen from the air, if It’s heavy, it won’t produce oxygen.
Under normal circumstances, the use of molecular sieve time can reach 1.5 to 2 years, in this time period, if found behind the oxygen machine white powder, at the same time there is intermittent drum-like sound on the need to replace the molecular sieve. Because the strength of molecular sieve itself is not particularly high, so when there is a white powder, it reflects that the wear degree of molecular sieve is relatively high, if not replaced in time, it will affect the effect of use, reduce the efficiency of oxygen production. Open the container and pour it out, as shown below, pour the molecular sieve in, fill it up, seal it, run it for two to seven days, open it again, add a little more, fill it up.
How to prolong the life of molecular sieve of oxygen generator? Do the following 10 points, can extend the life of medical molecular sieve oxygen machine.
- The medical molecular sieve oxygen machine should be placed smoothly, otherwise it will increase the noise of the medical molecular sieve oxygen machine.
- In the process of transportation and storage, it is strictly prohibited to put horizontally, upside down, wet or direct sunlight.
- Do not turn on the machine without the filter and screen, do not put the flowmeter ball on the zero position when the medical molecular sieve oxygen machine is on.
- The water level in the humidification bottle should not be too high (the water level should be half of the bottle body) , otherwise the water in the bottle is easy to escape or enter the oxygen inhalation tube.
- Medical molecular sieve oxygen machine work, as far as possible in the indoor cleaning position, from the wall or other objects not less than 20 cm distance.
- Clean up the medical molecular sieve oxygen machine, filter, or change the insurance, remember to cut off the power supply.
- The use of medical molecular sieve oxygen machine to avoid open fire, to avoid the occurrence of fire.
- Patients use medical molecular sieve oxygen machine, in case of power failure, such as the impact of patients with oxygen accidents, please prepare other emergency measures.
- When the medical molecular sieve oxygen machine is not in use for a long time, please cut off the power supply, empty the water in the humidifying bottle, wipe the surface of the medical molecular sieve oxygen machine, cover it with a plastic cover, and store it in a dry place without sunlight exposure.
- Pay special attention when filling the oxygen bag with the medical molecular sieve oxygen machine. After the oxygen bag is filled, you must first pull out the tube of the oxygen bag, and then turn off the switch of the medical molecular sieve oxygen machine, otherwise, it is easy to cause the negative pressure back-suction of the humidified bottle water into the medical molecular sieve oxygen generator, resulting in the failure of the medical molecular sieve oxygen generator.